The gas fluidic flowmeter of new shape with wide measuring range has been developed. The major point of this development is to widen the measuring range keeping pressure loss at maximum flow rate within 15mmH//2O. The design incorporates a venturi shaped and a wide feedback loop. The measuring range has been increased to approximately 80, and the Reynolds number at minimum flow rate with oscillation is decreased to approximately 21.
|Number of pages||6|
|Publication status||Published - 1985 Dec 1|
ASJC Scopus subject areas