We hypothesized that the externalizing direction of the driveline (the driveline angle) at the percutaneous exit site would influence the occurrence of driveline infection after left ventricular assist device implantation. From August 2013 to May 2017, 71 patients were implanted with a HeartMate II device in our center. The driveline angle was measured on anteroposterior radiography just after implantation. Risk factors for driveline infection were analyzed by uni- and multivariate analyses. Driveline infection developed in 10 (14%) patients during follow-up. Overall actual freedoms from driveline infection at 6, 12, and 24 months were 96%, 88%, and 86%, respectively. Overall number of driveline infection events per patient-year was 0.16. Receiver operating characteristic analysis determined the cut-off point of the driveline angle as 41°. The 6-, 12-, and 24-month actuarial freedoms from driveline infection in those with driveline angle more than 42° (84%, 74%, and 74%, respectively) were significantly lower than in those with driveline angle less than 41° (97%, 94%, and 90%, respectively; p < 0.02). The numbers of driveline infection events per patient-year were 0.16 in patients with driveline angle more than 42°, and 0.04 in patients with driveline angle less than 41°. Multivariate analysis demonstrated that driveline angle more than 42° was an independent risk factor for driveline infection (hazard ratio 4.71). Driveline angle more than 42° is an independent risk factor for driveline infection in patients with HeartMate II. Externalization of the driveline toward the horizontal direction is important to prevent driveline infection with HeartMate II.
ASJC Scopus subject areas
- Medicine (miscellaneous)
- Biomedical Engineering
- Cardiology and Cardiovascular Medicine