### 抄録

The method presented by Liu et al. in 1998 has generally been used in Japan to calculate the wedge factor of the enhanced dynamic wedge (EDW). When the points used to evaluate the dose exist outside the central beam axis in the half field (HF), however, a difference of about 3% can occur between the values calculated by this method and the measured values. In 2003, Liu et al. presented a new general formula for calculating MU using the EDW. We modified the formula for calculating the wedge factor by combining the conventional calculation formula and the formula of Liu et al., presented in 2003, and compared the calculated values of the wedge factor of the EDW in the HF with the measured values. Our formula employs a very simple method in which only the normalized golden segmented treatment table (NGSTT) on the center of the radiation field and the central beam axis are added to the existing formula. The values calculated by our formula and the measured values were consistent within 1% in most combinations of energy, wedge angle, and size of radiation field. When the EDW was used for irradiation to the tangent line of the HF breast, in particular, the values calculated by our formula and the measured values were consistent within 0.5%, and it was possible to calculate the wedge factor with a higher degree of accuracy than that calculated with the existing formula.

元の言語 | English |
---|---|

ページ（範囲） | 409-416 |

ページ数 | 8 |

ジャーナル | Nippon Hoshasen Gijutsu Gakkai zasshi |

巻 | 62 |

発行部数 | 3 |

出版物ステータス | Published - 2006 |

外部発表 | Yes |

### Fingerprint

### これを引用

*Nippon Hoshasen Gijutsu Gakkai zasshi*,

*62*(3), 409-416.

**Examination of formula for enhanced dynamic wedge factors in half fields.** / Sato, Tomoharu; Hanada, Takashi; Ito, Yasushi; Sasaki, Miwa; Ootsuka, Tsugio.

研究成果: Article

*Nippon Hoshasen Gijutsu Gakkai zasshi*, 巻. 62, 番号 3, pp. 409-416.

}

TY - JOUR

T1 - Examination of formula for enhanced dynamic wedge factors in half fields

AU - Sato, Tomoharu

AU - Hanada, Takashi

AU - Ito, Yasushi

AU - Sasaki, Miwa

AU - Ootsuka, Tsugio

PY - 2006

Y1 - 2006

N2 - The method presented by Liu et al. in 1998 has generally been used in Japan to calculate the wedge factor of the enhanced dynamic wedge (EDW). When the points used to evaluate the dose exist outside the central beam axis in the half field (HF), however, a difference of about 3% can occur between the values calculated by this method and the measured values. In 2003, Liu et al. presented a new general formula for calculating MU using the EDW. We modified the formula for calculating the wedge factor by combining the conventional calculation formula and the formula of Liu et al., presented in 2003, and compared the calculated values of the wedge factor of the EDW in the HF with the measured values. Our formula employs a very simple method in which only the normalized golden segmented treatment table (NGSTT) on the center of the radiation field and the central beam axis are added to the existing formula. The values calculated by our formula and the measured values were consistent within 1% in most combinations of energy, wedge angle, and size of radiation field. When the EDW was used for irradiation to the tangent line of the HF breast, in particular, the values calculated by our formula and the measured values were consistent within 0.5%, and it was possible to calculate the wedge factor with a higher degree of accuracy than that calculated with the existing formula.

AB - The method presented by Liu et al. in 1998 has generally been used in Japan to calculate the wedge factor of the enhanced dynamic wedge (EDW). When the points used to evaluate the dose exist outside the central beam axis in the half field (HF), however, a difference of about 3% can occur between the values calculated by this method and the measured values. In 2003, Liu et al. presented a new general formula for calculating MU using the EDW. We modified the formula for calculating the wedge factor by combining the conventional calculation formula and the formula of Liu et al., presented in 2003, and compared the calculated values of the wedge factor of the EDW in the HF with the measured values. Our formula employs a very simple method in which only the normalized golden segmented treatment table (NGSTT) on the center of the radiation field and the central beam axis are added to the existing formula. The values calculated by our formula and the measured values were consistent within 1% in most combinations of energy, wedge angle, and size of radiation field. When the EDW was used for irradiation to the tangent line of the HF breast, in particular, the values calculated by our formula and the measured values were consistent within 0.5%, and it was possible to calculate the wedge factor with a higher degree of accuracy than that calculated with the existing formula.

UR - http://www.scopus.com/inward/record.url?scp=33646841346&partnerID=8YFLogxK

UR - http://www.scopus.com/inward/citedby.url?scp=33646841346&partnerID=8YFLogxK

M3 - Article

C2 - 16604046

AN - SCOPUS:33646841346

VL - 62

SP - 409

EP - 416

JO - Nippon Hoshasen Gijutsu Gakkai zasshi

JF - Nippon Hoshasen Gijutsu Gakkai zasshi

SN - 0369-4305

IS - 3

ER -