The internal pipe wall of magnetic flowmeters should be nonconductive to prevent generated electromotive force from short circuiting. Usually the inside of metallic pipe is lined by insulating material. The lining limits applicable temperature range of measured fluid and also reliability. A new structure is proposed, in which the insulating liner is eliminated. A potential distribution is formed on the pipe wall by applying voltage proportional to fluid flowrate. The potential distribution is kept almost identical to the flow induced potential in the fluid so that no current flows across the boundary between fluid and wall. Therefore the output signal is exactly the same as that of conventional magnetic flowmeters.
|ホスト出版物のタイトル||Unknown Host Publication Title|
|出版者||North-Holland Publ Co|
|出版物ステータス||Published - 1984 12 1|
ASJC Scopus subject areas